This is not new news…this has been known for years, and locked down in later releases of the models impacted.
Its the same vulnerability that has existed, way back, to older models? Its what makes open APS work? Someone from industry got super mad at me for bringing it up, but he was a pacemaker guy. I told him that this stuff was well known for years. I guess I’m just looking for clarification from the diabetic community.
All our devices transmit plain text, right? Are you saying that maybe the 670 is different than earlier models? Nothing pops up for the 670g. Is that one different somehow?