Just saw that this one case is meant for use with wired through devices, but they have a warning about insulin pumps.
Also, here is a site that says that these are “watertight” not “waterproof”.
Below is a quote from the webpage:
Special Note: Regarding the use of AQUAPAC® cases with Insulin Pumps
Aquapac have commissioned research by Imperial College London into the use of the 150 wire-thru case after receiving a large number of customer enquiries regarding their suitability for use with insulin pumps. Results show that, in some circumstances, insulin flow rate can be reduced by the case. In particular, low flow rates are impeded, and as such Aquapac can make no claim regarding the suitability of these cases with insulin pumps. Use of these cases for insulin pumps is entirely at the discretion of the user. Aquapac has been advised to recommend that users of Aquapac cases with insulin pumps regularly check their blood sugar levels.
The tests were conducted by Imperial College of Science, Technology and Medicine using a Roche Accu-Check Spirit insulin pump at a temperature of 20C at flow rates of 2U/hr and 10U/hr.
The amounts of insulin delivered over 60 minutes observed were as follows:
2U/hr not in a case - 0.0131g (flow rate of 100%)
2U/hr in Aquapac 150 - 0.0106g (flow rate of 81%)
2U/hr in Aquapac 150 - 0.0056g (2nd test model) (flow rate of 42%)
10U/hr not in a case - 0.0864g (flow rate of 100%)
10U/hr in Aquapac 150 - 0.0878g (flow rate of 100%)
21st August 2006